Since Fukushima, much interest has developed in the application of checking food and water for possible radiation contamination. Here are your options:
In the field of radiation detectors, the two most popular designs are Scintillation Counters and Geiger Counters. Many scintillation counters are more sensitive in general, and able to detect certain radiation at greater ranges. These devices tend to be more expensive than Geiger counters, and generally larger, more specialized, and less compact in design.
The most popular design of radiation detector is a Geiger counter because it is readily available, easy to use, of compact design in many cases, and in an affordable price range. And certain models are quite sensitive. So for one who chooses to use a Geiger counter to check food and drink for radioactive contamination, certain criteria are recommended:
Of the Geiger counter models within our selection that meet at least 1 of the above criteria, here is a comparison:
Now for the specific details of checking for contamination, and here is my disclaimer - these are only procedures that I would follow, and are not represented as the best or most thorough, or foolproof.
In the end, the effectiveness of a Geiger counter in detecting contaminated food and drink, in my opinion, comes down to a matter of degree. Heavily contaminated food is potentially detectable by many Geiger Counters of reasonable sensitivity. Weakly radioactive food might be detectable by some pancake GM tube models in combination with a timed count process. But there could theoretically be a minimum amount of radiation particles missed by any Geiger counter. So anyone using a Geiger counter for this purpose needs to take responsibility for that decision, versus other alternatives.
As a matter of interest, I have scanned everything from milk to soy sauce to sake, and the only radioactive item I have found so far is a batch of captured Arizona rainwater. Specifically, I detected 6 CPM (Counts per Minute) of radiation from the sample, using the Digilert 100 Digital Geiger counter, a standard-tubed (not pancake) model, so that instrument is still pretty sensitive to have achieved that feat. I would add, though, that a momentary scan itself of the captured rainwater failed to reveal any contamination - the radioactivity become apparent only through a 20 minute timed count.
As an additional resource on the topic of detecting radiation in food, I repeat here a summary written by International Medcom, manufacturer of the Inspector Alert and Radalert 100 models of Geiger counter:
Accurate measurement of radiation in food requires a multi-channel analyzer and a special oven for ashing the food to concentrate the radioactivity. Our instruments have been used for experimental, educational, and screening purposes in checking food.
Measuring radiation in food is tricky. Naturally occurring radiation in potassium-rich food (such as bananas, when dried into banana chips, and salt substitutes) from Potassium 40 can easily be detected with the Inspector and (with less sensitivity) the Radalert 50. In the case of fallout from nuclear testing or accidents (such as Chernobyl), you would be looking mostly for Strontium 90, Cesium 137, and possibly Plutonium 239. Of our instruments, the Inspector is the best for this application because of its higher sensitivity. The Inspector’s efficiency for Sr90 and Cs137 beta is good, and it does detect the Cs137 gamma. It does detect Pu239, but Pu239 can have health effects at very low concentrations, which can be difficult to detect with any instrument.
Before you screen for radiation in food, you should establish a baseline measurement in the same location where you plan to test the food. It is best to accumulate the baseline counts for 12 hours as described below.
When you measure, you should put the mica window of the instrument directly over the food you are measuring, as close as possible. In the case of milk or other liquids, fill a container very close to the top so you can measure directly, without the glass in the way. If all your milk is from the same source, you might want to boil or evaporate some to concentrate it, then take a measurement from that. Set the display to Total and accumulate the counts for 12 hours in each location. Divide the total count for the period by the exact number of minutes to get the average CPM.
Back to Applications